Double Tilt LN2 Cooling -170°C Atmos Defend + Vacuum Transfer TEM holder
LN2 Cooling -170°C + Double Tilt±20° + Vacuum Transfer TEM holder
Get the High Resolution image and EDS Analysis at-170°C
Temp. Control ±0.005°C
High Resolution at -170°C data movie and data Collection
Sample： Au nano particle ARM 200F in Kyushu University Factory check resolution image
Sample： Au nano particle ARM 200F
in Kyushu University
Factory check resolution image
University of Tokyo
Dr. Ishikawa Dr. Nakayama
Prof. Shibata Prof. Ikuhara
Atomic resolution STEM image observation of LLZ(Li7La3Zr2O12),
which is degraded by exposure to air(moisture) and weak to
electron beam with lithium ion material, was successful.
From this result, it was proved that the non-exposure(Transfer)
was successful, and the crystal orientation was adjusted by Y tilt,
and the electron beam damage was reduced by cooling of -160°C.
Reasons for high-resolution image at -170°C
We realized it with the following our special system.
- Mechanical drift cancellation mechanism from our patented technology
- High-precision temperature controller . Range is ± 0.005 ° C .
- Original gonio cover for cryo tem holder
By making these into one system, it is possible to obtain much easyer get the high resolution image than current cooling holders.
As a result, the stability during cooling has improved, so we have successfully acquired EDS column map data.
EDS analytical performance test using LN2 cooling holder: Column mapping of SrTiO3
TEM holder Specification
||Double-tilt ,LN2 Cooling and VacuumTransfer
|Y tilt / Min step
||: ±20°/ 0.05°
||: Smart Set System
||: replenish LN2 repeatedly
Temp. controller, Y tilt controller,
software, Safety Operation Stand of tem holder introduce.
Safety Operation Stand of TEM holder